General Notes
===================
-If you receive the error "unable to enumerate amplifiers" or "unable to load okFrontPanel.dll", try installing .NET 3.5 via Microsoft Update or Windows Update.


Release Notes
===================

v0.84
-----------
- Boost settings are now linked to Rf Gain settings
- Boost is enabled by default on Pico2 for 3.3Gig and 10Gig settings
- Default settings can be over-ridden via the boost configuration file
- Format for Pico2 boost over-ride file (pico2_0.boost)
    - First line is comment line
    - Second line is number of Rf gains (default 5)
    - Subsequent lines (one per gain) are SPACE delimited with 3 fields
         Enable (0 or 1), Frequency (0 to 255), Gain (0 to 255)
         Example:
            pico2_0.boost (or any other comment)
            5
            0 128 128
            0 128 128
            0 128 128
            1 22 52
            1 70 178
- Format for Pico2 or Triton Duo gain over-ride file (pico2_0.cfg or triton_duo_0.cfg)
    - First line is comment line
    - Second line is number of Rf gains (5 for Pico, 4 for Triton Duo)
    - Subsequent lines (one per gain) are TAB delimited with 2 fields
         Display string, Rf value in exponent form
         Example:
            pico2_0.cfg (or any other comment)
            5
            10Meg	10e6
            100Meg	100e6
            1Gig	1e9
            3.3Gig	3.3e9
            10Gig	10e9

v0.83
-----------
- Fix Pico AutoCal problem on Windows XP
- For Pico, offset the displayed Cfast value, so approximately 0pF is
  displayed after compensating source=head with nothing attached
- Speed up AutoCal

v0.82
-----------
- Fix sources for Pico2

v0.81
-----------
- Config file support added to Pico2
- Bug fixes for Pico2

v0.80
-----------
- TritonDuo: add I=0 and Iclamp offset calibration
- TritonDuo: full Vclamp/Iclamp support completed

v0.79
-----------
- TritonDuo: config file support
- TritonDuo: combine I=0 and Iclamp into single mode.
  Use IclampOn bit to switch between the two modes.

v0.78
-----------
- Add Iclamp On/Off support for custom amplifiers

v0.77
-----------
- Add dual-register source select support to enable disconnecting
  the head, and allow Imodel to be decoupled from Iclamp circuit.

v0.76
-----------
- Add support for multiple Iclamp gain values for Tritons that
  support Iclamp.  Available gain values are 10x, 30x, 120x,
  and 250x.

v0.75
-----------
- For Picos and Tritons that support Iclamp (current clamp)
  mode(s).  Allow connecting/disconnecting of Iclamp circuit
  in Iclamp modes to achieve true I=0 in any of the Iclamp
  modes.  In the future, I=0 mode will be phased out, since
  any Iclamp mode can be used in I=0 mode.

v0.74
-----------
- Support for both Vclamp and Iclamp on new Triton amplifiers
- Enabling 100% Rs compensation on Pico is more reliable

v0.73
-----------
- Pico's I=0 & Iclamp gains are now 1x, 10x, 40x, 100x, 250x
- Preliminary support for TEVC Pico

v0.72
-----------
- Support new higher bandwidth Triton+
- Fix Pico's JP range to be +/-255mV (was +/-127mV)
- Eliminate register write delays on multi-channel amplifiers

v0.69
-----------
- Enable boost on Pico 2

v0.68
-----------
- Support Pico 2 gains without having to auto-calibrate scale

v0.64
-----------
- Fix Iclamp and I=0 modes for Triton
- Sync version numbers of TecellaLab and TecellaAmp SDK/API

v0.51
-----------
- Use C:\TecellaLabData for captures and protocols folders
- AutoScale down to Rf of 30Meg

v0.50
-----------
- AutoScale all Rf gains

v0.49
-----------
- Fixes Pico Iclamp offsets
- Can choose V1 or V2 GUI

v0.47
-----------
- Fixes crash on New Protocol

v0.44
-----------
- Fixes crash on save protocol

v0.43
-----------
- Fixes Pico Rs Comp bug

v0.42b
-----------
- Pre-release for 0.43
- Fixes performance issues

v0.42
-----------
- Fixes bug for TecellaLab version 1 save tlc file if no "captures" folder present
- Save file location cannot be changed in version 1
- Save file name is yyyyMMdd_hhmmss_zzz.tlc in version 1

v0.41
-----------
- Adds support for additional amplifiers.

v0.40
-----------
- Adds support for additional amplifiers.
- Keyboard shortcuts added for graph navigation.


v0.39
-----------
- Adds support for additional amplifiers.


v0.38
-----------
- Adds support for additional amplifiers.
- Adds level-histogram feature.
- Fixes units for ABF conversion.


v0.37
-----------
- Holding level for Iclamp can be set now.
- Stimulus editor no longer does weird things after it is closed.


v0.36
-----------
- Fixes buffer overflows for Jet and Richmond when using Zap.
- Fixed Advance step scan button. 
- Allows larger ranges for the DSP filters.


v0.35
-----------
- Step Scan - beta version added. 


v0.34
-----------
- Adds Iclamp enable/disable per channel if it's supported by the amplifier.
- Reorganize channels settings so auto compensations are always shown.


v0.33
-----------
- Fixes the "Use Stimulus" control.


v0.32
-----------
- Adds support for current clamp modes.
- Adds controls to synchronize acquisition with a trigger in.
- Implements simple file playback.


v0.31
-----------
- Enables a new offset adjust mode that simultaneously adjusts the stimulus (JP) and the response.


v0.30
-----------
- Fixes crash in TLC conversion from v0.29.


v0.29
-----------
- Fixes support for Turbo mode.
- Delta segments types have been fixed, but are handled differently than before.
	- The resulting stimulus format is not compatible with the old stimulus format.
- Program options are now saved in the user's directory and transfer from version to version.
- Fixed small things about stimulus programming and tlc conversion.
- Changed flow for how to use TecellaLab.
- Added unit prefixes to lots of places.


v0.28
-----------
- Added slew rate control on a per stimulus segment basis.
- Added digital output controls on a per stimulus segment basis.


v0.27
-----------
- Add slew rate control and labels.


v0.26
-----------
- In the middle of a lot of changes.
- TLC conversion added.
- Amplifier pairing doesn't work in this version.
- TLC viewing functionality temporarily disabled removed.


v0.25
-----------
- Added basic TLC viewing functionality.


v0.24
-----------
- Added frequency boost filter.
- Added auto zero and auto offset adjust functionality.
- Filter performance improvements.


v0.23
-----------
- TLC files record correctly again.
- Improvements made to auto scale.
- Uses TecellaTLC v0.3, which is now open source.


v0.22 (Deprecated)
-----------
- WARNING: Records corrupt TLC files.
- Holding voltage range was hard coded to +-255mV, now it changes depending on the actual range of the amplifier.


v0.21 (Deprecated)
-----------
- WARNING: Records corrupt TLC files.
- Supports new amplifier configurations.
- Fixes some spurious device contention errors, however others remain.


v0.20
-----------
- First release to use the new Opal Kelly drivers v3.1.0.
 Zap duration can be controlled via a toggle button.
- Added low pass filtering functionality.
- Auto calibration and auto scale are run automatically at startup.


v0.19
-----------
- Adds RMS statistics.
- Delay between repeats takes effect immediately, instead of at next repeat.
- Digital leak feature added.

v0.18
-----------
- Implements a delay between repeats.
- Prevents/handles conflicting amplifier commands more gracefully.

v0.17
-----------
- Fixes a bug where using multiple amplifiers would crash when running Zap.

v0.16
-----------
This version introduces a new GUI with dockable windows.
New features include:
	- Selectable amplifier configurations, if supported by the amplifier.
	- Automatically combines multiple connected amplifiers so it looks like one.
	- Automatic compensations now work during repeat acquisition.
	- Improved capacitance compensation algorithm.
	- Auto calibrate/scale/artifact can now be toggled on/off.
	- Added labels to the graph so it is easier to identify the channels.
	- Stimulus can be made relative to the holding voltage.
	- Artifact removal now displays it's equivalent capacitance.
	- The Statistics window updates periodically instead of after every run.
	- Improved graphing CPU usage.
